Tinytag Explorer, Tinytag Explorer Connect and Tinytag DI have a
calibration reminder feature that will prompt a user when a data
logger is due to be calibrated.
The default setting for a calibration reminder is 30 days before
a data logger's previous calibration is a year old, but this can be
changed in the software. The calibration reminder can also warn
users that they are using a data logger that has never been
To configure the settings for calibration reminders in your
Tinytag software, go to Options and Calibration Options. From
there, you will be able to:
- Set the date for a calibration reminder
- Turn off calibration reminders
Tinytag data loggers can be returned for a service calibration
at any time, where, in addition to the calibration, the unit's
batteries and any seals or desiccant packs will be replaced. We
recommend that all Tinytag data loggers have their batteries
changed once a year. The calibration interval should be determined
by the user according to the application that the data loggers are
used in; for the majority of applications, it is suitable to have
loggers calibrated annually.
For more information on Gemini Data Loggers' Calibration
service, please see our article on
what is included in the Tinytag calibration service.